Each Fluke product is warranted to be free from defects in material and The A Manual Set provides complete information for operators and service or. Warning If the A Calibrator is operated in any way not specified by the A Operators Manual or other documentation provided by Fluke, protection. View and Download FLUKE A operator’s manual online. Multi-Product Calibrator. A Test Equipment pdf manual download.
|Published (Last):||6 April 2007|
|PDF File Size:||20.49 Mb|
|ePub File Size:||16.93 Mb|
|Price:||Free* [*Free Regsitration Required]|
Set the communications parameters to match that of the A. Programming The Esr And Ese 15 errors that have occurred. Leveled Sine Wave Flatness Calibration. The verification procedures in this section provide traceable results; however the factory uses different procedures and instruments of higher precision than those described here.
Page The minimum use specifications are usually determined by maintaining a specified test uncertainty ratio between the calibration equipment and the unit under test. To access calibration steps described in paragraphs throughsimply send the command: Page Current Post command Select the active A Amplifier or A Calibrator front panel binding posts terminals for current output.
Dangerous voltages may be present on the leads and connectors. With the A Calibrator, you can calibrate precision multimeters that measure ac or dc voltage, ac or dc current, ac or dc power, resistance, capacitance, and temperature. Figure shows the arrangement of assemblies inside the A.
There are two zeroing functions: High Frequency Verification This procedure provides an example of testing high frequency flatness using a 5. Then follow these steps to for each period listed in Table Overload Function Verification This procedure uses the following equipment: The HPA digital multimeter can be programmed from either the front panel or over the remote interface to make these measurements.
Remove all the analog modules. Page 52 A Operator Manual You may operate the calibrator during warmup, but specifications are not guaranteed until warmup is complete. Set the filter on the PM as indicated in the flke.
Also see the ERR? Each status change register has an associated mask register.
Trigger Signal Specifications Edge Function. Applies to both simulated thermocouple output and thermocouple measurement.
Table of Contents Add to my manuals Add. See Table Interface Clear A control line that sets the interface to a quiescent state.
In addition to the settings shown below, adjust the scan control for a well-triggered display. For example, press the kilo multiplier key. How to Contact Fluke To contact Fluke, call one of the following telephone numbers: As soon as the spur is minimized, stop rotating Fluks.
Multiply the rms reading by the conversion factor listed to convert it to the peak-to- peak value. Connections for Four-Wire Compensated Capacitance.
Enter text from picture: The Event Status Register is a two-byte register in which the higher eight bits are always 0, and the lower eight bits represent various conditions of the A Calibrator. The 55000a harmonic will fluctuate, but there is a point at which both harmonics will be at the correct decibel level. The A prompts you to remove all cables from the front panel outputs.
Then follow these steps to calibrate pulse width: A dc reference voltage is supplied to the A51 board from the A6 DDS board; all dc and ac oscilloscope output voltages are derived from this signal and generated on the A51 board. Set the oscilloscope to look at the first 10 ns of the edge signal with the rising edge at the left edge of the oscilloscope display. Press the CAL softkey, opening the msnual activity menu below.
To register your product, visit register. Establish the flluke MHz reference: